The CSAT3 at 1m at station 1 is not working properly. I am quite
sure it was generating good data after I powered up the station today,
but within the first hour EVE no longer reported data from it.
The CSAT3 command shows all NAs, and no csat data is coming
I tried several things, none of
power cycled the serializer several times
power cycled eve
power cycled the serializer again
power cycled the sonic
blew out some moisture from the front panel connector
If you do a talk to it from eve, it is transmitting binary
gibberish as it should, but somehow eve is not recognizing the
records. It is as if it isn't sending the end-of-record bytes.
Using these commands:
ET= 10 ts=i XD=d GN=443a TK=1 P=5 FK=0 RN=1 IT=1 DR=102 rx=2 fx=038 BX=0 AH=1 AT=0 RS=1 BR=0 RI=1 GO=00000 HA=0 6X=3 3X=2 PD=2 SD=0 ?d sa=1
WM=o ar=0 ZZ=0 DC=6 ELo=020 020 020 ELb=020 020 020 TNo=898 d TNb=998 JD= 007
C0o=-2-2-2 C0b=-2-2-2 RC=0 tlo=8 8 8 tlb=8 8 8 DTR=01740 CA=1 TD= duty=028 AQ= 10 AC=1 CD=0 SR=1 X=0 MX=0 DT=02320 DTC=01160 RD=o ss=1 XP=2 RF=018 DS=007 SN0538 25oct01 HF=005 JC=3 CB=3 MD=5 DF=05000 RNA=1 rev 3.0a cs=10382 &=0 os=
The only thing I know to look for is RS=1, which is correct.
To restart it, did:
It started spewing data, but EVE is not recognizing it.